Measurements will begin when specified triggering conditions are met, following the receipt of the INIT command. Also shows how to enable and disable the SYNC output. A basic program that selects the waveshape and sets its frequency, amplitude, and offset. The example programs are provided solely for the purpose of illustrating a subset of the SCPI Standard Commands for Programmable Instruments commands used with the A. If set to -1, all points in memory will be transfered. Data will be cleared from instrument memory after transfer is complete.
|Date Added:||19 September 2007|
|File Size:||67.61 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
This can be executed at any time, including when the instrument is currently taking measurements.
Measurements will begin when specified triggering conditions are met, following the receipt of the INIT command. The example programs are provided solely for the purpose of illustrating a subset of the SCPI Standard Commands for Programmable Instruments commands used with the A.
Agilent — InstrumentKit Library documentation
Data is transfered from the instrument in bit double floating point precision format. IO You will need to acquire the environment and the IO independent of this download. Very easy to interface. Downloads an arbitrary waveform to the generator as binary data. Configures a pulse with duty cycle which is then slowly modulated by a triangle waveform. There is also a separate zip file for those programs that are published in the A User’s Guide.
They are organized into different zip files, depending upon the programming environment:.
If there are instgument data points available, the value 9. They are not to demonstrate the development environment or the IO. Legal Privacy Terms Trademark Acknowledgements. Immediately after the trigger conditions are met, the data will be sent to the output buffer of the instrument. Creates a linear sweep for a sinusoid.
33210A, 33220A Function / Arbitrary Waveform Generator IVI and MATLAB Instrument Drivers
The value True corresponds to the output being on, while False is the output being off. Modifies the parameters for a square wave, including duty cycle, and outputs a instrumnt of three cycles upon receipt of each trigger. Symmetry represents the amount of time per 33220z that the ramp wave is rising unless polarity is inverted.
Read the Docs v: They are organized into different zip files, depending upon the programming environment: This function allows the instrument to compensate of the voltage divider 3220a accurately report the voltage across the attached load. Configures a waveform with amplitude modulation using lower-level SCPI commands. Please don’t tag with vba questions that have nothing to do with VBA. The edge time is incremented. Sign up or log in Sign up using Google. Here are all the examples provided by Keysight: Also shows how to enable and disable the SYNC output.
You cna probably do a little research on your own for a start “Agilent A visual basic” as a web search? Post as a guest Name.
The latest version can be downloaded from: If currently taking a reading, the instrument will wait until it is complete before executing this command. First data point sent to output buffer is the oldest. For faster downloads, use BinaryArb.